KEYENCE LS-7601 PDF
Precautions for CE Markings Keyence confirms that the LS Series . LS- Photoelectric Sensor or Proximity Sensor Timing input signals are sent. Find great deals for 1 PC KEYENCE Ls CCD Laser Micrometer. Shop with confidence on eBay!. LS High-performance controller. Visible measurement pointController with Target Viewer. High-durability CCD . The LS Series measures the outer diameter of a fiber continuously and .. E-mail: [email protected]
|Published (Last):||26 January 2013|
|PDF File Size:||19.34 Mb|
|ePub File Size:||17.82 Mb|
|Price:||Free* [*Free Regsitration Required]|
The maximum value during the sampling period specified by external timing input is displayed and output as shown below.
Optimum Measurement Methods Refer to the following methods, and select the optimum measurement method according to the type of target. Average Hold The average value during the period specified is displayed and output. Refer to Calibration Type Selection on page For the electrical specifications of signals, refer to the reference pages indicated in the table. Setting Range The range is initially set to — The following example shows positive edge numbers and measuring portions.
Bottom Hold The minimum keyejce during the period specified is displayed and output. ComiXology Thousands of Digital Comics. The LS Series completes statistical sampling and starts statistical processing.
This range varies with the display unit or minimum display unit kryence.
The measurement of the object in two directions ensures high accuracy. Lywey g x 0. When repeating the two-point calibration, the previous settings will appear on the screen, which will be different from those in the above setting example. Page 18 Chapter 1 Getting Started System Applications The LS Series has a wide range of applications when used in combination with commercially available peripheral devices.
Page 8 Optimum Measurement Methods Refer to the following methods, and select the optimum measurement method according to the type of target. Control Panel Functions and Nomenclature Refer to page for RSC environment setting details.
If the number of edges detected does not coincide with the set value, the measured value will be invalidated. See pageThe measured value displayed drifts.
All function keys other than the display selection and tolerance setting function keys are locked. Then repeat the calibration settings again. The remote control console does not operate. Learn more about Amazon Prime.
See pageCheck that the target is not leaned or Locate the target correctly in the shifted. Environment Setting Record Sheet Target Viewer Function on page Tolerance display Displays the preset tolerance values. Refer to Section 4 Operation Control on page If the offset value is set, it is possible to display the offset value immediately on the screen by using the auto zero function.
See and discover other items: Amazon Drive Cloud storage from Amazon. These settings can be registered as programs that can be used whenever required. Chapter 10 Troubleshooting This section provides information on troubleshooting and the contents of error messages.
KEYENCE LS : Manual
The following applications are typical examples of measurement in a variety of Measuring modes. Page 13 Appendix List of Optional Accessories Please make sure that the end users are provided with this manual.
Ieyence following format will be used for the corresponding measured value. Page 71 Chapter 5 Function Settings Area Settings Area settings are used to determine the measurements taken from a variety of objects placed in the beam.
List of Optional Accessories Comparator waiting status with “—” displayed for each digit: The focus monitor range is the distance between the receiver and target in the measuring area. Drill Function The drill function can measure the external diameters of objects, such as drills or gears with odd-numbered teeth.
KEYENCE LS-7600 User Manual
Chapter 3 Easy Setting Guide Statistical Processing In the following example, the values calculated from the measuring data are: Page 46 Keynce 3 Easy Setting Guide Measurement of Outer Diameter and Eccentricity In the following example, the outer diameter and eccentricity of the roller is measured while the roller moves.
Normal The measured value is displayed and output as it occurs.
Chapter 5 Function Settings Calibration Settings The measured value of an target may have a delicate error due to the surface condition and angle of the target.