ASTM A578 PDF
ASTM A – pdf – Download as PDF File .pdf), Text File .txt) or read online. ASTM A/AM Standard Specification for Straight-Beam Ultrasonic Examination of Rolled Steel Plates for Special Applications. ASTM A/AM: Standard Specification for Straight-Beam Ultrasonic Examination of Rolled Steel Plates for Special Applications.
|Published (Last):||26 August 2007|
|PDF File Size:||2.89 Mb|
|ePub File Size:||14.16 Mb|
|Price:||Free* [*Free Regsitration Required]|
The American Society for Testing and Materials takes no position respecting the validity of any patent rights asserted in connection with any item mentioned in this standard. The method will detect internal discontinuities parallel to the rolled surfaces. Repeat the operation to establish the boundary. The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other.
Current edition approved Sept. Maintain this instrument setting during evaluation of the discontinuity condition. Combining values from the two systems may result in nonconformance with the speci?
Grid scanning shall be conducted along a continuous perpendicular line on the center. Referenced Documents purchase separately The documents listed below are referenced within the subject standard but are not provided as part of the standard. The Navic is capable of weld scanning girth welds and long seam weldsautomated corrosion mapping, and tank scanning.
If you feel that your comments have not received a fair hearing you should make your views known to the ASTM Committee on Standards, at the address shown below. Download Now White Paper: Make and model of instrument, test frequency, surface condition, transducer type and frequencyand couplant. Digital Transformation may be defined in a number of different ways by analysts or enterprise software vendors. The method will detect internal discontinuities parallel to the rolled surfaces.
The Largest Open Access Portal of Nondestructive Testing (NDT)
Supplementary requirements are provided for a587 of clad plate and for alternative procedures. Acceptance Standard—Level A 7. Historical Version s – view previous versions of standard Translated Version s: Use a suitable adtm such as water, soluble oil, or glycerin. The manufacturer may reserve the right to discuss rejectable ultrasonically examined material with the purchaser, the object being the possible repair of the ultrasonically indicated defect before rejection of the plate.
Students Click Here Join Us! Register now while it’s still free!
Defense manufacturing is all about project manufacturing and project accounting. If the back re? Referring to specific customer projects several of these cases are examined to outline the advantages of using Laser Shearography for automated defect detection.
The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Your comments will receive careful consideration at a meeting of the responsible technical committee, which you may attend. A curve is then plotted showing the deviations from the above established 2: The Eng-Tips staff will check this out and take appropriate action. Three levels of acceptance standards are provided. When details of these requirements are not covered herein, they are subject to agreement between the manufacturer and the purchaser.
This procedure covers the standard and acceptance standards for straight-beam, pulse-echo, ultrasonic examination of rolled carbon and alloy steel plates.
Use the transducers at their rated frequency. About Advertise Contact Login. The test shall be performed either by direct contact, immersion, or liquid column coupling.
Additionally, prior approval must be obtained if the weld repaired surface exceeds 1. All discontinuities causing complete loss of reflection shall be recorded. Close this window and log in. Acceptance Standard—Level C 9.
The surface of the reference block shall be no better to the unaided eye than the plate surface to be examined. It’s easy to join and it’s free. Click Here to join Eng-Tips and talk with other members!
ASTM A – acceptance standard clarification
The supplement number shall be added for each supplementary requirement ordered. aastm
Just to start you off in your standards selection review exercise: